The Boat Anchor Manual Archive Path: /hp/a/. This is the BAMA archive. These manuals are available for download and free of charge. I am now accepting. View and Download HP A operating and service manual online. DIGITAL VOLTMETER. A Measuring Instruments pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual.
|Published (Last):||28 July 2005|
|PDF File Size:||8.91 Mb|
|ePub File Size:||20.3 Mb|
|Price:||Free* [*Free Regsitration Required]|
Keysight / Agilent Digital Voltmeter – In Stock, We Buy Sell Repair, Price Quote
Learn more – opens in a manaul window or tab. This section of the manual contains the performance specifications and general operating characteristics of the 34SSA. The ac input attenuator is an RC circuit which provides a fixed attenuation of to-l on the V ac and V ac ranges. The input mea- surement is taken at this time.
Be sure to maintain null when the Reference Dividers output is changed. The A reading should be 0. High to Low Input Terminals: After calibra- tion, it should be left on and, if possible, kept in a con- trolled environment where the ,anual temperature is within one or two degrees of the temperature in which it was calibrated.
This voltage drop is proportional to the current through the resistor. This mode allows the measurement value to be modified by the addition, subtraction, multiplication or division of a known value. A service kit -hp- Part No.
This completes the dc common-mode Rejection test. B 66 Syntax Error – Indicates improper 1 program code. This condition shows up as being “One reading behind” in your measurement sequence. Alternating or direct current power line.
Interface Connections and Bus Mnaual. Remote Permits selected devices to be set to remote operation, allowing parameters and device cKaracteris- tics to be controlled by Bus Messages. For example, if manjal calibration chart indicates that the A output is 0.
Refer to the Operating Manual of your controller for in- structions.
Agilent / HP 3455A Digital Voltmeter
Hewlett-Packard uses a two-section serial number consisting of a four-digit prefix and a five-digit suffix separated manhal a letter manula the country in which the instrument was manufactured. Simplified Output Circuit The supplement is a duplication of the first three sections of this manual and should be kept with the instrument for use by the operator.
The A normally leaves the factory with the switch set to listen address 6 and talk address V decimal code Miscellaneous Troubleshooting Schematic 3 8-C Adjust the Test Oscillator level controls for a 34S5A reading as close as possible to the reading recorded in Step i.
The A display reading should be between 4.
HP 3455A Manuals
Remove the original AlORl 10 Figure The A 1 V. This section contains information for ordering replace- ment parts. After reassembly the meter is working just as well bp it did before.
The Test Cards may be reproduced without written permission from Hewlett-Packard. Front Panel Operation The Require Service message can be cleared by serial polling the A or by clearing the A. The Ohms Converter is a voltage limited current source which supplies a constant current through the unknown and reference resistors until the output voltage reaches approximately 4.
The talk and listen address is a 7-bit code which is selected to provide a unique address manal each bus instrument. Any ex- tra triggers sent during this cycle will be ignored.
Hewlett Packard HP a 6-digit Digital Voltmeter | eBay
As the example points out, there are regions of ambiguity even when the standard is ten times more accurate than the instrument being tested. The Self-Test mahual can be remotely programm- ed, as described in the programming portion of this sec- tion. The cal constant measurement can be aborted a number of times, depending on the function of the instrument.
During the offset error measurement all input switches to the inte- grator are opened. Shipping – 12 kg 26 tbs. Inguard Controller Reset Circuit High Resolution Mode For warranty service or repair, this product must be returned to a service facility designated by -hp.
Figure illustrates the circuit configuration for this measurement. The A requires periodic verification of perfor- mance. S ohm 1 kilohm – [5.