Download FLUKE PM PM PM PM 2X2MV,MHZ OSCILLOSOPE SM service manual & repair info for electronics experts. Re: Fluke PM manual needed (MHz scope) The service manual for Fluke / Philips PM PM PM PM can be found. PM ). Deflection Manual positioning: Horizontal, Vertical, Both . Web access: com.
|Published (Last):||19 August 2010|
|PDF File Size:||16.11 Mb|
|ePub File Size:||7.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
You need to be registered and approved to bid at this auction.
This diagram shows the final intensity Z and focusing amplifiers. Here, the accuracy of this range is checked.
The timing capacitance consists of C that is always in circuit and Tluke, C that are activated via the switching transistors V and V TV pattern generator with video output PM test set-up: CHIP A range of much finer steps can also be selected. Really don’t know why the mains fuse went. P,3082 Smith Hampton reserve the right to stop clearance if in their opinion, they believe it is being carried out in an unsafe manner or without the Method Statement and Risk Assessment.
Please note time changed to end from To reduce the rise of mmanual shock donot perform any servicing other than that contained in the operating See also section 9. Item Ordering code Description Mains input connector Fuse 5×20 1,6AT Contact spring single Battery compartment cover Page Probe Adjust signal is distortion-free and has equal intensity.
Remove all superfluous soldering material.
Oscilloscope Fluke PM3082 powersupply problem?
Anyway, it appears that I’ve got another nice looking four channel oscilloscope on my bench! Properties expressed in numerical values without tolerances, ranges or limits stated, represent the characteristics of an average instrument.
A rotary incorporates a rotor contact that can make contact with one of the two stator contacts. Manyal not present install 2 penlight LR6 back up batteries in the flukke at the rear panel of the oscilloscope.
These lots will be specified in the sale catalogue.
Damage to Premises The Rluke shall be responsible for the removal of all goods purchased at his own expense and risk and such removal must be carried out safely and lawfully and in accordance with any conditions of sale of which the Buyer is notified. Page – Maximum phase shift between horizontal a Recommended Test Equipment Note: Page Page – Horizontal deflection; X deflection coef Also for this the oscilloscope must be sent to the nearest Fluke Service Center.
Oscilloscope Fluke PM powersupply problem?
The oscilloscope must be capable of displaying signal amplitudes that are larger than the screen. Over and under-voltage protection To protect the ffluke circuitry against over-voltage and the supply circuits against overload, the power supply is provided ,anual a protection circuit.
Could an incorrect measurement setup cause the mains fuse to explode? Your question has been sent to the auctioneer. Biasing current for these outputs is provided via V, V and R Overseas purchasers will be required to provide details of: For installation, reverse the sequence.
If it was an AC power line fault of some sort lightning hit, lost neutral, hi-v distribution line falls on power line dropit would have destroyed other pieces of test equipment in the lab. I’ve never seen a shorted secondary side make the mains fuse explode. In the signal name list you find the itemnumber of the component that is source or destination.
Fluke PM3082 Service Manual
This is present on all 4 channels. Manusl the Auctioneers nor the Vendor shall be liable to account to the Buyer in the event of a re-sale at a higher price than the price contracted to be paid by the Buyer. This gives a time-linear voltage across the capacitor; the so- called sawtooth or sweep signal. Put a new one 1. The mains fuse has really exploded. That short may have blown open during the event, typically a copper track vaporises or a semi blows open.
This is displayed on the CRT with a message: No further dluke will be available after this time.
FLUKE PM SERVICE MANUAL Pdf Download.
Mabual digital multimeter and oscilloscope are not required for this test. Input Characteristics Note 1: If any written notice is delivered by hand, it shall be treated as having been received at that time.
Enter text from picture: Where stated on the lot description, the purchaser will be required to sign an undertaking prior to removal from site confirming their agreement to ensure all goods are tested and certified in accordance with the appropriate Regulations prior to use.
The linearity of the vertical amplifier is checked by moving a signal with a fixed amplitude vertically over the entire screen area.
The error code lists are a usefull help in faultfinding; in many cases a displayed error code will point to the defective circuit part. A brief examination of the primary side of the smps reveals no shorts. Verify that the delay difference between the two displayed signals is less than 0. The unit A1 description is split into the following mankal After this, the scope was left for several years